Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. Message 5 of Message 1 of 4. I have prepared two draft VIs numbered 4 , so as to save time. If you take a look at the block diagram, there is a Case Structure which checks to see if an error has already occured. The error you are seeing only occurs if there was an error before you called this VI. This abstracts the instrument functions to those in defined classes though I have not checked to see if IVI drivers are available for your instruments.
|Date Added:||19 November 2014|
|File Size:||52.78 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
I cannot run the examples in the library of the instrument.
Instrument Control (GPIB, Serial, VISA, IVI)
This is where the state machine begins to shine. Proud to say labviee I’ve progressed beyond knowing just enough to be dangerous. I have prepared two draft VIs numbered 4so as to save time.
Message 8 of Message 10 of The error you are seeing only occurs if there was an error before you called this VI. Message labvew of If an error occurs do the HP VIs execute? Message 3 of 4.
Getting GPIB Error Wait for Interruppt – Discussion Forums – National Instruments
So, there were two possibilities on a specific piece of hardware one cost 10, beads the other beads and you saved the customer and my. Thank you for your reply. I replace it lqbview the unlocked below. Since I am not regular programmer, it took me 2 days to do that.
The program should be the same except for the leakage current which isn’t in the initial sensing version.
Solved: issues about hpb driver – Discussion Forums – National Instruments
Jeff explains cost vs price. If what I said is correct basically you can do it but you need a very deep knowledge of what you are doing because each driver is made for one specific hardware.
Message 10 of Message 1 of The program up to this point only shows the I-V curve once it labvies finished the execution, and I’d need to see it “live”, that’s why I wanted to show the graph from labvkew sub-VI in the main VI. That is something you should probbaly learn, but may be more than you wish to tackle right now. Message 4 of If it has, then it reports the error that you are seeing.
Agilent Technologies / Keysight Technologies hp4140b Power Supply
I notice that you have set it so that it will attempt to set the output to zero when the loop is abotu to stop, whether due to reaching the limit or due to an error. This abstracts the instrument functions to hp4410b in defined classes though I have not checked to see if IVI drivers are available for your instruments.
Nonetheless, I am still not able to turn off the voltage after the shut-off current is reached, and so the applied voltage would still be applied and the sample would be damaged. Message 2 of Let me add something else. Message 2 of I’ve set up the required configuration for this combined measurement and it works properly no remote control-without labview.
The problem, as I explained in previous email, is that I would need to do something similar to what Mysticfree explains in the aforementioned discussion http: Message 1 of 4.
Message 6 of NI provides many and vendors provide them as well. My support system ensures that I don’t look totally incompetent.