NET, Service Pack 2 is required. The new version of the driver has been tested to be backwards compatible with previous versions. This operation actually runs: The supported model numbers are: If you need to go back to an older version of the driver, you need to uninstall the later version and install the older version. Message 6 of
|Date Added:||25 January 2005|
|File Size:||35.13 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
You can download the parser and its associated SDK from: I have my MAX setup as in the attached doc. I am trying to 3498a0 Agilent’s A mux unit 2. The simulation capabilities are designed primarily to support NI Switch Executive version 2. The easiest way to to install these other components is by running the Ivi Shared Components installation package available from agikent IVI Foundation as noted below. Message 9 of The is our engineering workhorse, we probably have 40 of them in the lab and use them in our production ATE systems.
If not found, the installer will either abort, warn, or install the required component as appropriate. Message 10 of The new interfaces agilet introduced rather than modifying the existing interfaces for backwards compatibility.
Agilent | Agilent A IVI-COM and IVI-C Driver Read Me
If you see an error about wscript. It is also available from: The simulation capabilities are designed primarily to support NI Switch Executive version 2. NET users with applications that error due to the situation above will need to change the declaration to the single line declaration shown above.
But I do have a lot of experience with the Agilent and it’s various modules, including the MUX. I am trying to simulate Agilent’s A mux unit.
Ivii how do i know which one is used? Supported Instruments Frame: Windows Installer Version 2. Is that the correct way of specifying the channel list ,?
It is able to install many of the required system components, including Windows Installer 2. This driver supports instruments manufactured by Agilent Technologies.
IviDmm init returns -1074134949 = File not found.
This driver supports instruments manufactured by Agilent Technologies. NET, Service Pack 2 is required. The problem is the way we declared the variables that are used to instantiate the driver object, and the fact that we later revised the COM interface of the driver.
Login Sign Up Ask a question. So the used DLL should be correct. Message 5 of I might be able to point you in the right direction. NET, Service Pack 2 is required. It contains “Getting Started” information on using the driver in a variety of programming environments as well as documentation on IVI and instrument specific methods and properties. More Information For more information about this driver and other instrument drivers and software available from Agilent Technologies visit: The problem was a defect module that reported a selftest error when turning instrument on.
See the section on installation in this document for information on what other software is required by this driver. For more examples on programming the instrument, please see the instrument Web page at www.